GE Logiq P5 Monitor showing NOT CONNECTED
December 18, 2018 09:37
I have a GE Logiq P5 when i give power monitor showing message NOT CONNECTED. Can someone guide me what is the problem?
to rate and post a comment
re: GE Logiq P5 Monitor showing NOT CONNECTED
December 19, 2018 09:03
Your monitor shows that message because has no signal, first open back cover of the monitor probably it will have a VGA or DVI connection unplug that connection and connect it to a laptop with a video cable,,, just to verify if the monitor is fine, if it is good,,, check video cable, pc inside the equipment.
re: re: GE Logiq P5 Monitor showing NOT CONNECTED
October 09, 2019 09:41
After checking the monitor,if its working properly,possible cause would be the video card or graphics card of the machine
Redstone Healthcare - Exceptionalism is what we do. Offering new and reconditioned ultrasound systems, transducers, parts and service nationwide from machines to transducers, C-arms and more! CALL 1-844-2-Redstone